Easy programming for high performance applications: Figure 12 Program Folder Item Options How To Use Auxiliary Functions My LabView program will sample the power value as the tunable laser is sweeping. Tunable Laser Sources How To Read Curve Values How To Remove Electrical Offsets
|Date Added:||4 September 2004|
|File Size:||48.94 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Attenuator Menu Options Table of contents Safety Considerations Also, the driver has built-in expert knowledge of the triggering and sampling behavior of the power meters, labviea relieves programmers from studying hardware details. Extended Trigger Configuration Agilent b Specifications Table 18 Overview For Supported Strings See search results instead: The Execute Lambda Scan Function Agilent a Specifications Thanks for the help and I’m too new in this area.
Auto Cal Off Currently, I’m trying to work out a labview program that sweep a wavelength range and log the wavelength and corresponding power value and save it to a file.
Furthermore, the driver is capable of stitching power measurement sweeps obtained in different power ranges for easily increasing the dynamic range of the resulting spectra.
I have found 0 info on this in the manuals or elsewhere. Setting Power Meter Parameters Could someone describe the correct procedure for measuring the receiver gain of an opto-electronic amplifier? How To Set Attenuation Setting Stability Parameter Adding A Connector Interface Line Power Connection I want to use the NB system agilenh characterize a polarization beam splitter.
How To Read Curve Values We are using NI F